Many SIGLENT products are RoHS compliant to EU 2015/863. This is also known as RoHS 3 Check the datasheet for the product in question to find more information.
Many SIGLENT SDS oscilloscopes, including the SDS6000A series, can perform automated Bode plotting to provide frequency response for power supply stability, filters, and more. This integrated approach simplifies testing and delivers results much more quickly than building your own system, integrating separate instruments with software and an external controlling computer. In the examples above, we … Continued
Starting in 2017, SIGLENT has been using this format for serial numbers for instrumentation:
Fuses are a common circuit element and are key to protecting from overvoltage and overcurrent situations that can damage important circuits. Fuse markings are typically stamped symbols located on the metal cap of the fuse. Here is a helpful guide to common fuse markings used with instrumentation, taking a common fuse type as an example: … Continued
Not at this time. Many instruments support communication via sockets (port 5025) and telnet (port 5024) but the port numbers cannot be changed. If you use VISA, you can implement LAN via TCP/IP, which does not need a port number. Here is a listing of current SIGLENT instruments with Socket and Telnet Support.
Simply access the Save/Recall menu for the instrument in question and select Local. The menu will show used/total amounts. Here, the SDG6X has approximately 83 MB and the SDG1X has 85 MB of memory:
The SDG series of arbitrary waveform generators allow you to create any function you wish. Here are the memory depths available for building arbs for each model:
If the SFDR is not listed directly in the specifications, you can find it by selecting the largest value between the harmonic and non-harmonic spurious data from the specification sheet at the frequency of interest. For example, here is the SDG6X generator specs for harmonic and non-harmonic distortion: If the signal of interest was … Continued
Per OSHA, a Nationally Recognized Testing Laboratory (NRTL) is an independent third-part laboratory recognized by the United States Occupational Safety and Health Administration (OSHA) to test and certify products to applicable product safety standards. Examples of NRTLs include UL, CSA, and TUV. Many SIGLENT products are TUV certified, which also covers UL and CSA requirements. To … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued