Currently listed datasheet calculation for SSA Frequency Uncertainty is being updated, but as of 08/26/2020, it has not. Here is the new calculation: We can use below new specification: ± (frequency indication × frequency reference uncertainty + 1/2(span/(sweep points-1)) + 10% × resolution bandwidth + marker resolution) For example: The frequency indication is 500 … Continued
the Insert a FAT32 formatted USB stick into the front panel Press FILE Set Browser to DIR and use the knob to select Local. The local drive is the internal storage of the instrument. U-Disk0 is the USB device. 4. Use knob to highlight the file you wish to copy (or press Operate and “select … Continued
The MAC address can be found on the LAN config interface. Path: System–interface–LAN.
The SIGLENT SSA3000X, SSA3000X Plus, and SVA1000X products all have an EMI option that adds a Quasi- Peak detector selection. The Quasi Peak detector is an integrating detector type that will automatically apply to all active traces on the analyzer. To prevent existing traces from being overwritten with the Quasi Peak detector data, please follow … Continued
If you have an antenna with calibration data, chances are it includes a number of columns of data. One column should be the frequency being tested. The others can include free-range performance, gain, and more. Here is an example of antenna correction data from a German antenna manufacturer: The most useful entry is the AF, … Continued
The SIGLENT SVA series of analyzer instruments are very flexible platforms that offer a number of different functions that can be activated. New instruments are delivered with over 100 hours of trial time for each option, allowing users to try the options before purchasing them. The trial timer is running while the instrument is powered … Continued
The SIGLENT SVA1X series is a multi-purpose RF measurement instrument. This operating tip shows you how to quickly configure the Vector Network mode to show a Smith chart displaying complex impedance: VNA mode – Press Mode > VNA Set Meas type as S11 or S21 Set Format – Smith Select R + … Continued
The SIGLENT USB-GPIB adaptor currently works with the following SIGLENT instruments: Multimeters – SDM3065X/SDM3065X-SC SDM3055/SDM3055-SC SDM3045X Function/Arbitrary waveform generators – SDG6000X Series SDG5000 Series SDG2000X Series SDG1000X Series SDG1000 Series Spectrum & vector network analyzers – SSA3000X Series SVA1000X Series Oscilloscopes – SDS1000CFL
Here is the original independent review of the SVA1015X FW V126.96.36.199.13 from Chris Grossman: Our development team was working on a new revision (V188.8.131.52.1) that included the following updates: New Functionality: Add CNR, Harmonic measurement in AMK options for SA Mode Add port extensions, user cal kit, new marker functions, correction switch, multi-traces and multi-format overlay display … Continued
SIGLENT instruments with USB ports can supply 500mA steady current at 5 V.