Not at this time. Many instruments support communication via sockets (port 5025) and telnet (port 5024) but the port numbers cannot be changed. If you use VISA, you can implement LAN via TCP/IP, which does not need a port number. Here is a listing of current SIGLENT instruments with Socket and Telnet Support.
Simply access the Save/Recall menu for the instrument in question and select Local. The menu will show used/total amounts. Here, the SDG6X has approximately 83 MB and the SDG1X has 85 MB of memory:
Yes. As long as the signal amplitude and frequency do not exceed the input specifications. When measuring ACV using the ACV mode, the DC components will be blocked. When measuring DCV using DCV mode, the AC components will not be blocked, but you can minimize the AC effects by using a slower test speed or … Continued
Per OSHA, a Nationally Recognized Testing Laboratory (NRTL) is an independent third-part laboratory recognized by the United States Occupational Safety and Health Administration (OSHA) to test and certify products to applicable product safety standards. Examples of NRTLs include UL, CSA, and TUV. Many SIGLENT products are TUV certified, which also covers UL and CSA requirements. To … Continued
Product Type Product Series Socket port 5025 (Y/N) Telnet port 5024(Y/N) Digital Oscilloscope SDS5000X Y Y SDS2000X Plus Y Y SDS2000X N N SDS2000X-E Y Y SDS1000X N N SDS1000X-E Y Y SDS1104X-U Y Y SDS1000CML+ N N SDS1000DL+ N N Waveform Generator SDG6000X Y Y SDG2000X Y Y SDG1000X Y Y SDG800 N N … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued
The voltage categories of SDM3045X, SDM3055, and SDM3065X DMMs is CATI 1000V and CATII 300V. The indication is located on the front panel label:
Yes. EasyDMM can support multiple instruments, but it can only actively measure on one instrument at-a-time. Simply add instruments using the Add Instruments interface. Here, we show 3 active SDMs connected via LAN: NOTE: Mode must be switched to Measure or Scan before data can be collected on the selected instrument.
SIGLENTs EasyDMM software provides easy instrument control and data collection. In this note, we will show how to configure EasyDMM for the first time. Download and install EasyDMM. You can find the download instructions and files here: DMM Software 2. Connect the SDM to the controlling computer or network. The SDMs have USB and LAN connections. … Continued
The SIGLENT SDM series of multimeters has a Power On Last function that recalls the last instrument configuration settings used previous to a power cycle. For example, if you have the meter in ACV mode with fixed range, and power cycle the instrument, it will power on in ACV mode with fixed range. The instrument … Continued