how to use custom sensor Some DMM Models added an ability to create a custom temperature sensor in a firmware update. Here is a summary for creating that sensor data on a SDM3055-SC 1. Turn on the temperature measurement function of the multimeter. 2. Turn on custom sensor options. 3. Create a sensor by yourself. … Continued
Many SIGLENT products are RoHS compliant to EU 2015/863. This is also known as RoHS 3 Check the datasheet for the product in question to find more information.
Starting in 2017, SIGLENT has been using this format for serial numbers for instrumentation:
Fuses are a common circuit element and are key to protecting from overvoltage and overcurrent situations that can damage important circuits. Fuse markings are typically stamped symbols located on the metal cap of the fuse. Here is a helpful guide to common fuse markings used with instrumentation, taking a common fuse type as an example: … Continued
The SDG800, SDG1000, SDG1000X, SDG2000X, SDG6000X and SDM3000 can return bitmap images of their displays by sending the following command: SCDP
Test Current Source: 1mA Max Resolution: 0.01Ω Input Protection: 1000V Input Protection Open-circuit Voltage: <8V Beep Threshold (short-circuit resistance): from 0 Ω to 2000 Ω
Not at this time. Many instruments support communication via sockets (port 5025) and telnet (port 5024) but the port numbers cannot be changed. If you use VISA, you can implement LAN via TCP/IP, which does not need a port number. Here is a listing of current SIGLENT instruments with Socket and Telnet Support.
Simply access the Save/Recall menu for the instrument in question and select Local. The menu will show used/total amounts. Here, the SDG6X has approximately 83 MB and the SDG1X has 85 MB of memory:
Yes. As long as the signal amplitude and frequency do not exceed the input specifications. When measuring ACV using the ACV mode, the DC components will be blocked. When measuring DCV using DCV mode, the AC components will not be blocked, but you can minimize the AC effects by using a slower test speed or … Continued
Per OSHA, a Nationally Recognized Testing Laboratory (NRTL) is an independent third-part laboratory recognized by the United States Occupational Safety and Health Administration (OSHA) to test and certify products to applicable product safety standards. Examples of NRTLs include UL, CSA, and TUV. Many SIGLENT products are TUV certified, which also covers UL and CSA requirements. To … Continued