The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
Information regarding product compliance is available on the datasheet for each product. Contact your local Siglent office for more details if you have trouble finding what you need. You can also write email@example.com
Test and measurement equipment is exempt from FCC emissions limits per CFR 47, Part 15, para 15.103(c): https://www.gpo.gov/fdsys/pkg/CFR-2002-title47-vol1/pdf/CFR-2002-title47-vol1-sec15-103.pdf Specifically: “(c) A digital device used exclusively as industrial, commercial, or medical test equipment.”
The SDM series of multimeters have autorange capabilities that allow the meter to automatically select the range appropriate for a given input signal level. To prevent damage to the meter, current (ACI and DCI measurement modes) ranges will be automatically up-ranged if the input signal exceeds the maximum value for the selected manual range. In … Continued
The SDM3000 series of Digital Multimeters feature dual measurement capability. This means that the instrument can alternate between two measurements and simultaneously display both values. Here are the functions that qualify for dual measurements: Here is an example of the display of an SDM measuring DCV and DCI using the dual measurement technique: … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
Press the Shift button and the 2nd button to go to the setting interface, then select the System Setup and then select the System Info to view the serial number and version number.
SIGLENT Technologies uses a broad range of products that utilize the USBTMC unified protocol to communicate with the host computer and control. Users can use SIGLENT provided PC software to send commands to remotely control the instrument. In addition, you can use NI-MAX (National Instruments Measurement and Automation Explorer) for instrument control. Here is an … Continued
Excerpt from the ANSI-ASQ National Accreditation Board: http://anab.org/news/newsletter/winter-2014/nist-traceability-statements NIST Traceability Statements Many calibration customers request on their accredited calibration certificates a statement that the measurement is traceable to the National Institute of Standards and Technology (NIST). They want assurance that the standards used for the calibration are ultimately traceable to the International System of Units … Continued
What instruments accept SCPI commands for remote control? All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.