There are 6 types of trigger status messages for the SIGLENT SDS series of oscilloscopes. The trigger status indication is located near the upper-right-hand side of the display, as indicated in this picture: Here are their definitions: Armed. The oscilloscope is acquiring pre-trigger data. All triggers are ignored in this state. Ready. All pre-trigger data … Continued
The SIGLENT USB-GPIB adaptor currently works with the following SIGLENT instruments: Multimeters – SDM3065X/SDM3065X-SC SDM3055/SDM3055-SC SDM3045X Function/Arbitrary waveform generators – SDG6000X Series SDG5000 Series SDG2000X Series SDG1000X Series SDG1000 Series Spectrum & vector network analyzers – SSA3000X Series SVA1000X Series Oscilloscopes – SDS1000CFL
SIGLENT instruments with USB ports can supply 500mA steady current at 5 V.
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
The term “div” for most SIGLENT specifications is equal to the Vertical Scale value (Commonly shown in units of Volts/div). Take the specification for the DC Gain Accuracy of a SIGLENT SDS1052DL+ For ranges ≤ 100 mV/div, the DC Gain Accuracy is given as ± [3% × (|reading|+|offset|) +1% × |offset| +0.2 * div+2 mV] If: Reading = 0.5 … Continued
Information regarding product compliance is available on the datasheet for each product. Contact your local Siglent office for more details if you have trouble finding what you need. You can also write email@example.com
Here is a list of probes currently supplied with each product: Click here to view the Product Probe List
The SDS series of SIGLENT Digital Oscilloscopes have a number of save options that include copies of the displayed image (Bitmap, JPG, and PNG are common image types) as well as instrument configurations (Setups) and data (CSV). Here are brief instructions on how to save files to external USB memory devices. Insert a USB memory … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
In some extreme hardware errors, the oscilloscope may become unresponsive or “hang” during the boot sequence. If your oscilloscope is frozen, will not boot, or appears to be unresponsive, you can attempt a hardware reset by following these instructions: 1. Disconnect all USB, LAN, and input cables. Leave the power cord connected. 2. Power cycle … Continued