The current version of the SPD programming commands (located in the Quick Start Guide) do not include any front panel lock commands. These are typically used to control the functionality of the front panel keys. Locking the front panel prevents accidental changes to instrument operation during remote control. The SPD1000X and SPD3303X/X-E have three commands … Continued
The SIGLENT USB-GPIB adaptor currently works with the following SIGLENT instruments: Multimeters – SDM3065X/SDM3065X-SC SDM3055/SDM3055-SC SDM3045X Function/Arbitrary waveform generators – SDG6000X Series SDG5000 Series SDG2000X Series SDG1000X Series SDG1000 Series Spectrum & vector network analyzers – SSA3000X Series SVA1000X Series Oscilloscopes – SDS1000CFL
SIGLENT instruments with USB ports can supply 500mA steady current at 5 V.
The SIGLENT SPD series of power supplies have only positive voltage and current setpoint values. That is, you can select voltages or currents that are from 0 to the max value (0-32 V, for voltage.. as an example for the SDP3000X/X-E). Here are the suggested wiring configurations to achieve both positive and negative potentials at the … Continued
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
Information regarding product compliance is available on the datasheet for each product. Contact your local Siglent office for more details if you have trouble finding what you need. You can also write email@example.com
The SIGLENT SPD series of power supplies offer constant current mode. This setting limits the current output of the specified channel and can be used as a constant current source and to protect your device-under-test (DUT). In this tip, we show the typical performance of the SPD as it switches between constant voltage (CV) and … Continued
The SPD series of power supplies feature remote programming capabilities and can be controlled via USB or LAN connection. In this note, we will use Python and PyVISA to create a program that will connect to an SPD, enable the output, delay, and then disable the output. There are two areas that differ from some … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
Linear DC regulated power supplies (“linear supplies”) alter the output voltage to the required value through the power frequency (50/60 Hz alternating current) voltage transformer. After being lowered (or raised) to the appropriate AC voltage value, the waveform is rectified, filtered, and stabilized by a voltage regulation circuit, and is then output as a regulated … Continued