In this video, we review some of the new enhancements included in the latest release (6.1.37R2, and OS rev V2) for the SIGLENT four-channel X-E oscilloscope series. We cover the new date/time stamp feature, Network Time Protocol, and datalogging.
This Video shows the overall capability of the new Siglent SDS1000X-U Series oscilloscope. The new model features a 100 MHz analog bandwidth with 4 analog input channels. The design leverages the popular SIGLENT X-E User Interface delivering a powerful and easy-to-use scope ideal for your signal measurement needs.
In this video, we use the Search and Navigate feature of the SIGLENT SDS2X Plus scope to isolate and indicate low voltage conditions of an input signal. Here, we use an SDG signal generator to produce a 60 Hz sine wave and we introduce a few periods of much lower voltage.. simulating dirty power/brownout conditions … Continued
In this video, I compare the response from two power supplies using a stepped load to show the differences between a well-tuned supply and one that exhibits excessive ringing. Also, add in some basic instrument requirements to help get you started making your own step load tests. Here, I use a SIGLENT SDL1X-E DC Electronic … Continued
The SIGLENT SDS1004X-E, SDS2000X-E series of oscilloscopes feature a Power On Main function that automatically turns the scope on when there is main power available. This is ideal for remote locations where it can be difficult to reset the instrument after a power outage.
The SIGLENT SDS1004X-E series (4 channel only. Does not apply to the SDS1202X-E), SDS2000X-E, and SDS5000X series of oscilloscopes feature a web control interface that provides remote instrument control without additional software. In this video, we show you how to configure the scope for wired LAN or wireless WiFi connection and an overview of the … Continued
The SDS1000X-E series of oscilloscopes feature adjustable zoom reference points for both vertical and horizontal scaling. Learn how this feature can help you troubleshoot more effectively.
This video explains the process of adding MSO/Digital input channels to the SIGLENT SDS1004X-E series of oscilloscopes.