Currently listed datasheet calculation for SSA Frequency Uncertainty is being updated, but as of 08/26/2020, it has not. Here is the new calculation: We can use below new specification: ± (frequency indication × frequency reference uncertainty + 1/2(span/(sweep points-1)) + 10% × resolution bandwidth + marker resolution) For example: The frequency indication is 500 … Continued
the Insert a FAT32 formatted USB stick into the front panel Press FILE Set Browser to DIR and use the knob to select Local. The local drive is the internal storage of the instrument. U-Disk0 is the USB device. 4. Use knob to highlight the file you wish to copy (or press Operate and “select … Continued
The MAC address can be found on the LAN config interface. Path: System–interface–LAN.
The SIGLENT SSA3000X, SSA3000X Plus, and SVA1000X products all have an EMI option that adds a Quasi- Peak detector selection. The Quasi Peak detector is an integrating detector type that will automatically apply to all active traces on the analyzer. To prevent existing traces from being overwritten with the Quasi Peak detector data, please follow … Continued
If you have an antenna with calibration data, chances are it includes a number of columns of data. One column should be the frequency being tested. The others can include free-range performance, gain, and more. Here is an example of antenna correction data from a German antenna manufacturer: The most useful entry is the AF, … Continued
The SIGLENT USB-GPIB adaptor currently works with the following SIGLENT instruments: Multimeters – SDM3065X/SDM3065X-SC SDM3055/SDM3055-SC SDM3045X Function/Arbitrary waveform generators – SDG6000X Series SDG5000 Series SDG2000X Series SDG1000X Series SDG1000 Series Spectrum & vector network analyzers – SSA3000X Series SVA1000X Series Oscilloscopes – SDS1000CFL
SIGLENT instruments with USB ports can supply 500mA steady current at 5 V.
EasySpectrum is free software used to control and collect data from SIGLENT SSA and SVA spectrum analyzer products. State files are used by the SIGLENT SSA and SVA instruments to recall or save specific instrument configuration settings. By storing and recalling states, you can quickly configure the analyzer for different tasks. To use State files … Continued
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
Information regarding product compliance is available on the datasheet for each product. Contact your local Siglent office for more details if you have trouble finding what you need. You can also write email@example.com