Currently listed datasheet calculation for SSA Frequency Uncertainty is being updated, but as of 08/26/2020, it has not. Here is the new calculation: We can use below new specification: ± (frequency indication × frequency reference uncertainty + 1/2(span/(sweep points-1)) + 10% × resolution bandwidth + marker resolution) For example: The frequency indication is 500 … Continued
the Insert a FAT32 formatted USB stick into the front panel Press FILE Set Browser to DIR and use the knob to select Local. The local drive is the internal storage of the instrument. U-Disk0 is the USB device. 4. Use knob to highlight the file you wish to copy (or press Operate and “select … Continued
The MAC address can be found on the LAN config interface. Path: System–interface–LAN.